Hydrocephalus simulator for testing of active ventriculoperitoneal shunts
Marianne Schmid Daners1, Vartan Kurtcuoglu2
1 Institute for Dynamic Systems and Control, Department of Mechanical and Process Engineering, ETH Zurich, 8092 Zurich, Switzerland
2 The Interface Group, Institute of Physiology, University of Zurich, 8057 Zürich, Switzerland
Duration: 2 years Project Completion: 2016
Background and Aim
Hydrocephalus is a pathological condition that is characterized by an active distension of the ventricular system, which is caused by an impaired flow through the cerebrospinal fluid (CSF)-pathways . Since CSF consists mainly of almost incompressible water, its accumulation causes an enlargement of the ventricles and, more importantly, can lead in the absence of invasive therapy to a degeneration of the brain parenchyma. Hydrocephalus is typically treated by diverting the excessive CSF from one of the two lateral ventricles via a shunt to other bodily compartments, such as the peritoneum (Figure 1). Although shunts are not a novel development, the typical outcome of this treatment is far from satisfactory. The deficiency is being addressed by the development of new, sophisticated active shunting systems in which the draining range can be adjusted to the actual need of the patient . However, before these devices can be implanted in patients, extensive testing and validation are required.
Current shunt-testing platforms are too simplistic to assess these new devices, in so far as they are incapable of replicating the interdependence between the shunt, the CSF-system and the intra-peritoneal dynamics . The deficiency will render necessary the undertaking of animal trials at all stages of the development process. With this project, we aim to provide a testing platform that can replace the preliminary animal studies. It reveals the fluidic dynamic functionality of a shunt in a realistic and accurate environment and reproducibly simulates the patient’s response to the device.
Method and Results
The simulator is based on the hardware-in-the-loop principle: the component to be tested is integrated into an artificial setup, with the usual environment being simulated in parallel. This was implemented by:
- using a posture mechanism to modify the location and orientation of the tested shunts according to a predefined test scenario;
A realistic testing environment is ensured by placing the shunt in a water bath and heating shunt and test medium to 37 ± 1 °C using a surrounding incubator. A lumped parameter model is used to simulate the patient’s relevant pathophysiology. This mathematical model includes the influences of the measured drainage rate, posture, cardiac-induced pulsations, and viscoelastic effects.
In two studies [5, 6], we have shown that the HIL testing platform can be used: to accurately analyze and quantify the dynamic interaction between shunt and patient; to test and evaluate anti-siphon devices in a reproducible yet realistic environment; and as a development platform for actively controlled valves.
Conclusions and Relevance for 3R
The developed in-vitro shunt-testing platform will render obsolete most animal trials during the first stages of the development process of future active CSF shunts. Its accessibility to the taking of measurements, as well as the repeatability and the reproducibility of the in-vitro tests, helps to define the questions that still need to be answered more precisely in animal models. But these experiments with animal models will be undertaken only when satisfactory results appertaining to the various functionalities have been achieved in the in-vitro simulations. Using this approach, the otherwise inevitable animal trials can be markedly reduced in number.
1. Rekate, H. L. (2008). The definition and classification of hydrocephalus: a personal recommendation to stimulate debate. Cerebrospinal Fluid Res, 5(2), 2.